With the new TRC-NW-400 non-mydriatic fundus camera Topcon enters a new area. An area where easy operation, an extremely small footprint, and superb images are key issues.
True color images
The Topcon TRC-NW400 produces true color fundus photos of excellent quality. Details are very clear and sharp.
The TRC-NW400 has the traditional 3 fixations for nasal, central, and temporal images. Besides that, you can photograph the complete periphery due to nine fixation targets for peripheral imaging. With TRC-NW400 you perform a quick screening or do a more elaborate examination.
You can acquire fundus images for small pupils, as small as 3.3mm. It means you can photograph nearly all customers. Due to the high sensitivity and low flash intensity you can produce excellent images with low flash and reduce patient’s stress.
In Stereo photography mode, the alignment for a stereo pair is performed automatically. Following the prompts, on-screen a stereo pair for stereo viewing can be quickly and easily acquired.
The Topcon TRC-NW400 has a very small footprint which minimizes the use of valuable space in a practice. A tiltable screen enables the eye care practitioner to use the fundus camera from several sides. Due to the flexible set up, there is more interaction with the customer and it saves valuable space. The Topcon TRC-NW400 can comfortably be operated when located against a wall or in a corner.
With just one finger touch the operator can take images of both eyes. The fully automated Topcon TRC-NW400 assures fully automated alignment, autofocus, and auto shoot.
The Topcon TRC-NW400 is DICOM compliant, making it easy to integrate within PACS and EMR programs.
|OBSERVATION & PHOTOGRAPHY OF THE FUNDUS:|
|Angle of coverage||45°/30° or equivalent (digital zoom)|
|Working distance||34.8mm (in fundus photography)|
|Pupil diameter for photography||45°, ø4.0mm or more|
|Small pupil diameter: ø3.3mm or more|
|Fixation target||Internal fixation target:|
|– DOT matrix type organic EL|
|– The display position can be changed and adjusted|
|– The displaying method can be changed|
|Peripheral fixation target:|
|– This is displayed according to the internal fixation|
|– target displayed position|
|External fixation target|
|PATIENT DIOPTER CORRECTION RANGE:|
|Without the diopter compensating lens*||-13D to +12D|
|When the concave compensation lens is used||-12D to -33D|
|When the convex compensation lens is used||+11D to +40D|
*Area where the split lines are used